A VALIDATED RP-HPLC METHOD FOR IMPURITY PROFILING OF SODIUM NITROPRUSSIDE IN INJECTION DOSAGE FORM
Objective: The main objective of this research work is to develop and validate a single reverse-phase high-performance liquid chromatography (RP-HPLC) method. This method should becapable of quantifying all the known, as well as other possible degradation impurities of sodium nitroprusside (SNP) in its injection formulation.
Methods: Of allmethod development trails, we have observed better separations between known and degradation impuritiesin Inert sustain C18, (250 x 4.6) mm, 5 µm column at 30 °C temperature. Isocratic elution was carried out by using pH 8.6 phosphate buffer and acetonitrile in the ratio of 65:35 %v/v with a flow rate of 0.8 ml/min. The detection was carried out at 220 nm, with an injection volume of 10 µl.
Results: In the proposed method, SNP was eluted at 22.5 min. Nitrite, nitrate, and ferrocyanide were linear from 0.25 to 37 μg/ml, ferricyanide was linear from 1.0 to 37 μg/ml, and SNP was linear from 0.75 to 37 μg/ml. The % RSD for six spiked samples (precision)was found to be less than 0.5 %. Accuracy was performed for known impurities from LOQ to 150 % for a 0.5 % specification level. The resultswere found to be in the acceptance range of 90-110 %. The LOQ concentration of nitrite, nitrate, and ferrocyanide was 0.25 μg/ml each,LOQ offerricyanide and SNP was found to be 1.0 μg/ml and 0.75 μg/ml, respectively. The SNP injection samples were exposed to different degradation conditions, and the results were found specific in the proposed methodology.
Conclusion: The proposed RP-HPLC method is specific, precise, accurate, linear, stable, and robust for quantification of known and other possible degradation impurities in SNP injection formulation.
1. Sodium Nitroprusside, USP Monographs, official (USP42-NF37), 4045.
2. Sodium Nitroprusside - European Pharmacopoeia 10.3, (01/2016: 0565).
3. NITROPRESS® (sodium nitroprusside injection) product information leaf let, Hospira, Inc., Lake Forest, IL 60045 USA, Document Name: QEN-3406v4.qxp, Reference ID: 3442071.
4. Taylor HT, Styles M, Lamming JA. Sodium nitroprusside as a hypotensive agent in general anaesthesia. br j anuesth 1970; 42: 859-64.
5. Moraca PP, Bitte ME, Hale ED, Wasmuth EC, Poutasse FE. Clinical evaluation of sodium nitroprusside as a hypotensive agent. Anesthesiology 1962; 23: 193-99.
6. Cottrell EJ, Casthely P, Brodie DJ, Patel K, Klein A, Turndorf H. Prevention of Nitroprusside-Induced cyanide toxicity with Hydroxocobalamin. N Engl J Med 1978; 298:809-11.
7. Sodium Nitroprusside, USP Monographs, official (USP42-NF37), 4045.
8. Sodium Nitroprusside - European Pharmacopoeia 10.3, (01/2016: 0565).
9. Gerritse GR. Rapid simultaneous determination of nitrate and nitrite by high-performance liquid chromatography using ultra violet detection. J Chromatogr A 1979; 171:527-29.
10. Chou SS, Chung JC, Hwang DF. A High Performance Liquid Chromatography method for determining Nitrate and Nitrite levels in vegetables. J Food Drug Anal 2003; 11:233-38.
11. Lim HS, Hwang JY, Choi EA, Lee G, Yoon SS, Kim MK. Development and validation of HPLC method for the determination of ferrocyanide ion in food grade salts. Food Chem 2018; 239:1167-74.
12. Frank JM, Johnsonx BJ, Rubin HS. Spectrophotometric determination of Sodium Nitroprusside and Its Photodegradation Products, J Pharm Sci 1976; 65: 45-8.
13. David MB, Marilyn DS, Nancy K and James EC: High-performance liquid chromatographic determination of sodium nitroprusside, J Chromatogr A 1981; 212:339-46.
14. Mahony C, Brown EJ, Stargel WW, Verghese PC, Bjornsson DT, In Vitro Stability of Sodium Nitroprusside Solutions for Intravenous Administration. J Pharm Sci 1984; 73:838-39.
15. Iram F, Iram H, Iqbal A, Husain A. Forced Degradation Studies. J Anal Pharm Res 2016; 3: 1-5.
16. Blessy M, Ruchi DP, Prajesh NP, Agrawal YK. Development of forced degradation and stability indicating studies of drugs-A review. J Pharm Anal 2014; 4:159-65.
17. Reynolds WD, Facchine LK, Mullaney FJ. Available guidance and best practices for conducting forced degradation studies. Pharm Technol 2002; 26: 48-56
18. International Conference on Harmonization (ICH) Harmonized Tripartite Guidelines, Validation of Analytical Procedures, Text, and Methodology, Q2 (R1), Parent Guidelines on Methodology Dated; 1996. p. 6.
19. David MB, Marilyn DS, Nancy K and James EC: High-performance liquid chromatographic determination of sodium nitroprusside. J Chromatogr A 1981; 212:339-46.
20. InertSearch TM for LC, Inertsil ® Applications, Analysis of Potassium ferricyanide and Potassium ferrocyanide, Data No. LB018-0811.
21. Huang HT, Salter G, Kahn LS, Gindt MY. Redox Titration of Ferricyanide to Ferrocyanide with Ascorbic Acid: Illustrating the Nernst Equation and Beer–Lambert Law. J Chem Educ 2007; 84:1461-63.
22. Jahanvee KT, Chirag JP, Patel MM. RP-HPLC method development and validation of macitentan with its known and unknown degradation impurities in its tablet dosage form. Int J App Pharm 2018; 10: 81-9.
23. Adison F, Sanjay Pai PN. A Validated stability indicating rp-hplc method for estimation of tolfenamic acid in presence of its pharmacopoeial impurities. Int J App Pharm 2019; 11: 264-270.
24. International Conference on Harmonization (ICH), Stability testing of new drug substances and products, Q1A (R2); 2003.
This work is licensed under a Creative Commons Attribution 4.0 International License.